Previous: Observing Projects
- because of an offset in the
phase lock loop which was not compensated for in the software, old
spectral correlator data are shifted by 97.656 kHz. Velocities must be
corrected downwards (blue-shifted) for the lower side band, upwards
(red-shifted) for the upper side band.
- Improved computer facilities in Grenoble allow
several data reductions to be carried out simultaneously.
Investigators are requested to contact S.Guilloteau to schedule the
data reduction in Grenoble. Because of unusual hardware problems
mentioned above, investigators should schedule a stay longer than
7 days in Grenoble for each project, with ample time for discussion
with IRAM astronomers.
- Users of CLIC who have a copy of the software at there
home institute are warned that the UV table creation was bugged in
the last release. They should contact S.Guilloteau or R.Lucas to
obtain an up-to-date version.
Thu Mar 9 14:51:14 MET 1995