Next: STORE CORRECTION
CLIC\STORE AMPLITUDE [/BAND code] [/SELF]
*** THIS HELP HAS NOT BEEN UPDATED FOR THE NGRX ***
Store the amplitude calibration determined by the last SOLVE AMPLITUDE.
One calibration curve is stored for each sideband. With option "/BAND
code", one may use the calibration curve determined with one of the
sidebands (`code' is UPPER or USB, LOWER or LSB, or AVERAGE or DSB), to
calibrate both sidebands.
With option "/SELF", the calibration stored will not be the calibration
curve (as determined from the last SOLVE AMPLITUDE command) but, for
each scan, the visibility corresponding to the first subband set (eg.,
SET SUB L01 to L03 ...); data from each side band will be used to cali-
brate the same side band. This provide a simple, easy way to self-cali-
brate data taken on strong continuum sources (quasars).
STORE AMPLITUDE and STORE PHASE can be performed simultaneously using
STORE AMPLITUDE PHASE [/BAND code] [/SELF] [/RECEIVER irec] (the /RE-
CEIVER option has no meaning for amplitudes).